Regular maintenance of medical endoscope equipment
Regular maintenance of medical endoscope equipment is generally carried out by instrument managers in collaboration with engineering technicians. It is an organized and planned maintenance measure that is continuously circulated, which is conducive to mastering the operating rules of the instrument and finding faults. The content and time of regular maintenance vary depending on the instrument. Generally, it can be divided into three levels:
(1) First maintenance: Generally, it can be carried out once a month to once a quarter. In addition to daily maintenance work in the factory, the main content can be to disassemble the casing, collect dust, dirt, foreign objects, tighten screws, and add lubricants in all places; Inspect the service components for wear, deformation, erosion, breakdown, looseness, moisture, aging, grounding, and other conditions; Detect the voltage and ripple of each group of power supplies, check the operation and contact conditions of high-voltage components, etc;
(2) Secondary insurance: Generally, it can be carried out once every six months to a year. The main content is not only to ensure the safety of the equipment, but also to test, calibrate, and calibrate the sensitivity and accuracy of various instruments and operating control systems on the control console of the entire machine, replace expired loss products such as high-voltage generator insulation, and conduct systematic testing and bias testing of the voltage and waveform at various test points in the circuit;
(3) Three guarantees: Generally, it can be carried out once every 2-4 years. In addition to ensuring the second guarantee, the entire machine can be fully disassembled for cleaning and maintenance if necessary. Components that exceed the service life should be replaced or repaired as much as possible. The instrument should be thoroughly debugged to restore its working accuracy and performance, and it is entirely possible to achieve or exceed the level of a new machine.
Cleaning and maintenance of medical endoscopes
The maintenance personnel of the endoscope equipment remind you that when operating medical endoscopes, the action should be gentle, the bending angle handle should be rotated slowly, and the force should be even. Do not use violence or excessive force. When the bending part is "fixed", do not turn the bending handle, otherwise it will break the traction steel wire . When holding a camera, it is necessary to firmly grasp the joint position of the camera and handle, and avoid grasping the wiring harness and hanging the joint position, otherwise it will damage the fiberglass bundle.
During the surgical process, the fiber optic endoscope instrument should be promptly wiped clean with sterile water after use, and the instrument should not be opened and soaked to avoid loss or damage to the body. If there are any other matters, please come to our website http://www.jmmoorephoto.com consulting service